Electronic fully automatic hydrostatic balance to determine Alcoholic Strength by Volume (ASV) on the distillate in 5 seconds, Total Dry Extract and Musts Evaluation.
The readout of the Alcoholic Strength by Volume is shown on a large graphic led screen simultaneously with the readout of temperature, density at 20°C and other parameters.
Sample holder cylinder with loading funnel and overflow, automated cylinder discharge device and motorized floater movement.
- Alcoholic strenght by volume (ASV)
Reading range 0,13÷99,98% vol
Readability 0,01% vol
Readability and reading repeatability ±0,03% vol (beetween 15 and 25°C)
Measurament range relative density 0 5÷2,25 (d20/20)
Readability and reading repeatability 0,00005 ± 0,00005
- Total Dry Extract
Range of values: 0,0÷505,8 g/l of sucrose
Readability and reading repeatability ±0,3 g/
- Musts evaluation (d20/20)
Relative density (d20/20) range 1,04÷1,16
Type Thermo probe in platinum Pt 100 1/3 DIN
Reading repeatability ±0,05°C (in the range 10÷30°C)
- Power pack,
- 2 hydro repellent floater interchangeable in mass and volume,
- certified hydroalcoholic solution ACCREDIA RMP,
- waste tank.
- Floaters with certificate,
- ACCREDIA RMP hydroalcoholic solutions (5÷60%ASV)
FLOATERS INTERCHANGEABLE IN MASS AND VOLUME
A special care is devoted to the construction and calibration of the floaters.
They are all interchangeable in mass and volume to simplify their use; for example, it is possible to use two different floaters to determine the relative density of a must or a wine; so that sample pollution is avoided.
In order to achieve what mentioned before, the suspension wire realized with a non-magnetic and stainless material, with a diameter less or equal to 0.20 mm. and the superficial treatment of the floater’s glass were object of thorough studies to nullify the problems caused by the surface tension, by the wire’s own volume, and by the electrostatic charges.
DATA SHEET SUPERALCOMAT ENG Rev 1.pdf